Low Speed Handpiece Maintenance Procedure
1: Clean the Chuck Mechanism.
A: Lubricate Chuck Mechanism with 1 drop of Lubricant.
(using Health Canada/FDA Approved Mineral Oil Lubricant)
B: Insert Chuck Brush into Chuck Mechanism.
C: Turn Chuck Brush in a circular motion, back and forth for 5 seconds.
D: Lubricate chuck mechism with 1 drop of Lubricant; repeating Step A.
E: Run Handpiece (without bur) for 5 seconds to expel debris.
Chuck Brush is Reusable - Please Ensure Suitable Cleaning of Chuck Brush after Every Use.
Chuck Brush and all other Maintenance Tools can be found in the New Handpieces tab, under Accessories.
2: Lubricate and Clean the Handpiece.
A: Disassemble Head from Contra Angle. (as shown in diagram below)
B: Wipe off parts with alcohol; never place any parts in an ultra sonic unit.
C: Place Lubricant (2 or 3 drops) into all parts noted by the arrow. (as shown in diagram above).
D: Reassemble the Handpiece and place it on a Slow Speed Motor and run for 30 seconds.
E: After Autoclave the Handpiece, allow to cool gradually for 30 minutes to room temperature.
F: Repeat Steps A through D.
Tips & Tricks
Make Sure your Handpiece is Well Lubricated.
Make Sure Autoclave Temperature is at Manufacturers Recommended Tempaterature (120c).
Do Not Subject your Handpiece to Any Heat above 132c/210f.
We Suggest to Maintain Your Autoclave.
Do Not Lubricate a Hot Handpiece.
We suggest Not To Reuse Burs.
Make Sure Chuck Mechanism & Handpiece is Cleaned and Lubricated After Every Use.