Low Speed Handpiece Maintenance Procedure
Clean the Chuck Mechanism.
A: Lubricate Chuck Mechanism with 1 drop of Lubricant.
(using Health Canada/FDA Approved Mineral Oil Lubricant)
B: Insert Chuck Brush into Chuck Mechanism.
C: Turn Chuck Brush in a circular motion, back and forth for 5 seconds.
D: Lubricate chuck mechanism with 1 drop of Lubricant; repeating Step A.
E: Run Handpiece (without bur) for 5 seconds to expel debris.
Chuck Brush is Reusable - Please Ensure Suitable Cleaning of Chuck Brush after Every Use.
Chuck Brush and all other Maintenance Tools can be found in the Our Products tab, under Accessories.
Lubricate and Clean the Handpiece.
A: Disassemble Head from Contra Angle. (as shown in diagram below)
B: Wipe off parts with alcohol; never place any parts in an ultra sonic unit.
C: Place Lubricant (2 or 3 drops) into all parts noted by the arrow. (as shown in diagram above).
D: Reassemble the Handpiece and place it on a Slow Speed Motor and run for 30 seconds.
E: After Autoclave the Handpiece, allow to cool gradually for 30 minutes to room temperature.
F: Repeat Steps A through D.
Tips & Tricks
Make Sure your Handpiece is Well Lubricated.
Make Sure Autoclave Temperature is at Manufacturers Recommended Temperature (120c).
Do Not Subject your Handpiece to Any Heat above 132c/210f.
We Suggest to Maintain Your Autoclave.
Do Not Lubricate a Hot Handpiece.
We suggest Not To Reuse Burs.
Make Sure Chuck Mechanism & Handpiece is Cleaned and Lubricated After Every Use.