High Speed Handpiece Maintenance Procedure
1: Clean the Chuck Mechanism.
A: Lubricate Chuck Mechanism with 1 drop of Lubricant.
(using Health Canada/FDA Approved Mineral Oil Lubricant)
B: Insert Chuck Brush into Chuck Mechanism.
C: Turn Chuck Brush in a circular motion, back and forth for 5 seconds.
D: Lubricate chuck mechism with 1 drop of Lubricant; repeating Step A.
E: Run Handpiece (without bur) for 5 seconds to expel debris.
Chuck Brush is Reusable - Please Ensure Suitable Cleaning of Chuck Brush after Every Use.
Chuck Brush and all other Maintenance Tools can be found in the Our Products tab, under Accessories.
2: Lubricate and Clean the Handpiece.
A: Remove handpiece from unit tubing, clean with alcohol.
B: With Bur inserted and the Coupler Attached, place Lubricant (2 or 3 drops) into the air drive port on the Coupler (as indicated in the photo below). At the same time, rotate the bur between your fingers, back and forth for a few seconds.
C: Re-connect handpiece to unit tubing, and run for 30 seconds to eliminate excess oil from the turbine.
D: After Autoclaving the handpiece, allow cool gradually for 30 minutes to room temperature.
E: Repeat Steps A through C.
Tips & Tricks
Make Sure your Handpiece is Well Lubricated.
Make Sure Autoclave Temperature is at Manufacturers Recommended Tempaterature (120c).
Do Not Subject your Handpiece to Any Heat above 132c/210f.
We Suggest to Maintain Your Autoclave.
Do Not Lubricate a Hot Handpiece.
We suggest Not To Reuse Burs.
Make Sure Chuck Mechanism & Handpiece is Cleaned and Lubricated After Every Use.