High Speed Handpiece Maintenance Procedure
Clean the Chuck Mechanism.
Step 1: Lubricate Chuck Mechanism with 1 drop of Lubricant.
(using Health Canada Approved Mineral Oil Lubricant)
Step 2: Insert Chuck Brush into Chuck Mechanism.
Step 3: Turn Chuck Brush in a Circular Motion, Back & Forth for 5 seconds.
Step 4: Lubricate chuck mechanism with 1 drop of Lubricant; Repeating Step 1.
Step 5: Run Handpiece (with Bur or Shipping Post) for 5 seconds to expel debris.
Chuck Brush's are Reusable.
Please Ensure Suitable Chuck Brush Cleaning after Every Use.
Lubricate & Clean the Handpiece.
Step 1: Remove handpiece from unit tubing, clean with Alcohol.
Step 2: Free Spin your Turbine: With Bur Inserted and Coupling Attached, place Lubricant (2 - 3 drops) into the air drive port on the Coupling (As indicated in the Diagram Below). At the same time, Rotate the Bur between your fingers, back and forth for a few seconds.
Step 3: Re-connect handpiece to unit tubing, and Run for 30 Seconds to eliminate excess oil from the turbine.
Step 4: After Autoclaving the handpiece, allow cool gradually for 30 Minutes to Room Temperature.
Step 5: Repeat Steps 2 & 3.
Tips & Tricks
Make Sure your Handpiece is Well Lubricated.
Make Sure Autoclave Temperature is at Manufacturers Recommended Temperature (120c).
Do Not Subject your Handpiece to Any Heat above 132c/210f.
We Suggest to Maintain Your Autoclave/Sterilizer Regularly.
Do Not Lubricate a Hot Handpiece.
We suggest Not To Reuse Burs.
Make Sure Chuck Mechanism & Handpiece is Cleaned and Lubricated After Every Use.
Check our Main Causes of Handpiece Failure for Details.